MICROCURRENT ELECTRICAL STIMULATION | DIGITAL EMS UNIT
Not Available in Canada*
3 Modes: Constant, Alternating, Synchronous
Pulse Width: Adjustable: 30-300 Microseconds
Timer: Adjustable from 1 to 60 minutes
Output Intensity: Adjustable, 0 to 60 mA (maximum output) into 500 ohm load each channel. Each channel is adjustable by a control knob.
Wave Form: Asymmetrical BiPhasic square pulse
System Includes: Stimulator, Lead Wires (New FDA compliant), 4 Self-stick Hypoallergenic Reuseable Electrodes, 9 Volt Battery, Hard Plastic Carrying Case, Instruction Booklet
*This medical device not licensed according to Canadian Law
(FDA) requires a prescription for the sale of electrotherapy devices to individuals in the U.S. (PRESCRIPTION LINK WORD FILE)
PROM-550 DIGITAL EMS FEATURES:
Channel I ON/OFF and Amplitude Control
Belt Clip (rotates 360° )
Channel II Output Jack
9 Volt Battery Compartment
Channel II ON/OFF and Amplitude Control
Channel I Output Jack
Function Control and Battery Compartment Cover
HOW TO USE THE PROM-550 DIGITAL EMS UNIT
1. Check Battery
Be sure you are using a fresh battery. Insert battery into battery compartment matching + and - indicators.
2. Prepare Skin
Before attaching electrodes, identify those placement areas where your physician has recommended. Each location should be cleaned, rinsed and thoroughly dried.
3. Lead Wires
4 Self-adhesive, reuseable electrodes are supplied with the unit. Replacements available in correct size as specified by the supplier ( 2 mm).
4. Insert Pin into Electrode Socket Insert pin connector into electrode socket. It is preferable to insert the pin connector before attaching the electrode to the skin.
5. Adjusting the Controls
Be sure both amplitude controls for Channel I and II B are turned to " OFF " position.
6. Connect Lead Wire
Plug end of lead wire into the channel output jack to be used (A and / or B), pushing plug in as far as it will go.
7. Place Electrodes on Skin
The use of a conductive medium is subject to the specifications of the component parts contained in the unit or recommended by the dealer.
Concerned about Replacement Lead Wires? The main consideration when buying replacement lead wires ("cables") for your EMS device is the connection size, such as 2.0 mm or 3.5 mm. All TENS and EMS devices can usually use the same types of lead wires, provided the connection size is correct for the device.
PROM-550 DIGITAL TENS UNIT TECHNICAL SPECIFICATIONS:
Dual, isolated between channels
Width X Length X Height:
6 X 10 X 2.5(cm) / 2.4 X 4 X 1(inches)
9 V DC battery
Adjustable, from 0 to 30V( Max. output), 500ohm load
Adjustable, from 0-60mA(Max. output) into 500phm load each channel. Each channel is adjustable by a control knob.
2 mm Connection
Adjustable, from 1 to 150 Hz
Adjustable, from 30 to 300µS
Asymmetrical biphasic rectangular pulse
Constant / Alternating / Synchronous
ON time, adjustable from 1 to 30 seconds
OFF time, adjustable from 1 to 30 seconds.
Width X Length X Height:
22 X 30 X 6.5(cm) / 8.8 X 12 X 2.61(inches)
Weight: 685 g / 1.51 lb
Main unit X 1, Battery X 1, Manual X 1, Electrode Pad X 4, Lead Wire X 2